13 Years Professional Beauty Equipment Manufacturing And OEM/ODM Experience
This instrument is the combination of the high-tech bio technologies and intelligent technologies, which chooses the distinguished and professional prescriptions, together with the unique combination concepts, and adopts the particular frequency-scanning electric circuit design according to the human body's biological theories and the electronic project theories. This program integrates with the weight-losing, body-beautifying, breast building, toxin draining and health-keeping. There are over 30 kinds of the compound computer prescriptions and 10 channels of impulse output, which is specially designed for the entire obesity ,local obesity ,drooping breast and hip, loose skin and other various aches. Its super combination of the frequency and make the nursing process more perfect and more professional.
◪ Dispel the accumulation fat which is in the waist, hip, belly and thigh, strengthen theactivity of the local muscle in order to shape the beauty of the body;
◪ Strengthen the organization cells, promote the metabolism ,dispel the waste and toxin of the metabolism;
◪ Dredge the tissue and venation, recycle the lymph and promote the circulation of blood;
◪ Relax the muscles ,ease the ache, drain the aches caused by the working fatigue andso on.
3,4: Work / Stop Indicator
5,6,7: Speed Shown (3 Models)
8: Error Warning Indicator
9: Timer Display
10,11: Time Up / Down
12: Program Select
13-16:Program Shown (Function1,2,3,4)
17: Auto Program
18: Auto Indicator
19: heat switch
1). Arrange the body-slimming procedures for the clients according to their
requirements and adjust the indoors temperature to keep the air fresh and ventilated before the operation; 2). Let the client have a bath or clean the parts that are needed to nurse ;
3). Connect the electrode flakes with the output lines, then insert the output lines into
the output sockets of the mainframe; 4). Fix the flex on the parts that are needed to nurse, but not tightly;
5). According to the conductor layout picture, draw the flex slightly and place the conductor covered with the medium into the flex, making the applicable side on the
skin and the distance of the conductor should be above 2~3 mm. 6). Recheck whether the flex is tight enough , the electrode flake is placed correctly ,touch
the skin tightly or the output conductive lines connect with the electrode flakes or not.
Notices in the operation
1). The intensity of the impulse output should be kept in the acceptable range for the clients when doing the operation;
2). Please press the START or STOP button to readjust when the conductive flake slips in the operation.
3). No matter which part of the body is in the operation, the working time had better not exceed 30 minutes. And if the same part of the body needs to do the operation again, the internal should be 30 minutes or 1 hours.
4). Not allow to use the erosive chemical liquid on the conducting side of the electrode flake. If it is stained with the accumulated liquid or unclean things, it will weaken the conductive capability .So the conductive flake should be cleaned often and kept clean.
5 .Not allow to place the conductive flake on the injured skin. If it needs to use ,it should be adhered with the porous adhesive bandages. And then place the conductive flake.
6). Please feel free to use the instrument if you feel slightly strained or swollen for the first time, for the lactic acid that produces by the metabolism of the body., which will disappear in a few days.
1). Before the operation, first to set 220/110V interchange button to fit your local voltage, (for example: local 110V voltage, set machine to 110V through adjust and set the 220/110v interchange button, against 220V, same set way.).
2). Connect the machine to your local voltage, turn on the machine, at this moment, the timer shown 30 minutes set.
3). According to operating demands, to adjust timer to a right work time. The max work time is 60 minutes. (if one of mix-max button or all doesn’t be off, the machine is to have warning voice, should turn off all mix-max buttons and wait 5 minutes, the warning voice stop).
4). Press select key to choose one program from Function 1 to Function 4, or press auto key and machines auto set the programs.
5). Press speed key to choose and get right output energy.
6).Insert the machine and put the electrode pads on the treating area, two different electrode pads in same group should put on body symmetrical or within 5 cm range, wrap it with bandage. (warning: always don’t put the paddle on heart area.).
7).Press start/pause key, ready to operate.
8).Adjust mix-max button to get suitable output energy on each path.
9).The piling of the infrared ray: adjust the button to select the power(small,middle, large) according to the need of the clients. adjust the button[20-29] to select the power(small,middle, large) The electrode flake will give off slight heat . Choose the power intensity according to the acceptance of the clients. Consult with the client frequently in order to avoid the injury by the over-heat.
◆ Make sure that the land-joining equipment and the creepage prevention switch is good.
◆ Not allow to place the electrode flake upon the heart of the human body less it will cause the uncomfortable of the client.
◆ The output intensity should be increased from small to large gradually. And consult with the client for their feelings in time. It is prohibited to accelerate or strengthen the intensity suddenly.
◆ If it needs to transfer the prescription, firstly, please press the PAUSE button and adjust the output intensity to the lowest. Reselect the prescription and adjust the suitable intensity output.
◆ The clients should have more water after the nursing for the convenience of draining the lymph toxin.
◆ Please care about the clients' feeling when output the infrared ray thermal energy for fear the over-heat injure the skin.
◆Please adjust output impulse after fix the electrode flake upon the body .
The following serious patients not allow to use the instrument: the one who have been equipped with the pacemaker; the one is with the heart failure, pregnancy , gastrolith, cystolith, kidney stone, etc . Technical Parameters
● Power voltage:AC220V/50HZ
● Power consumption:<100W
● Groups of the energy output:10
● Frequency output:5~1000HZ
The voltage output of the infrared ray
Electric current (MA)
Frequency range (Hz)
● 10pairs currents pads with infrared
● special electric wire with infrared
● one pcs breast enhancer currents pads
● 10 pcs bandages
● a big bandage for the waist
● one holder
● one power cord